# home-brew vswr meter

T

#### tom

Jan 1, 1970
0
How minimal of circuit would be required to tell the difference between
whether a load was inductively reactive or capacitively reactive at
frequencies between 100mhz and 200mhz, and in the 5 to 100 watts range? A
one, basically.

A

#### Active8

Jan 1, 1970
0
How minimal of circuit would be required to tell the difference between
whether a load was inductively reactive or capacitively reactive at
frequencies between 100mhz and 200mhz, and in the 5 to 100 watts range? A
one, basically.

A VSWR meter won't tell you that, just that there's a mismatch.
Start longer and prune for best VSWR. That's as simple as a couple
of diodes, a meter, and some resistors and/or a pot - maybe a power
range switch in leiu of the pot and a switch for forwared/reflected
power. Additionally you'd need some type of directional coupler - 2
parallel PCB traces... or a piece of insulated 24 AWG wire threaded
under a length of coax braid... or a torroidal coil with the coax
running through it. Unless you can get a crossed-needle display,
you'll have to calculate the VSWR from the fwd/rev readings or just
prune until the reflection are minimum. Knowing the VSWR would help
you compare with other antennas.

Oh, yeah. Is google's server down or something?

F

#### Fred Abse

Jan 1, 1970
0
How minimal of circuit would be required to tell the difference between
whether a load was inductively reactive or capacitively reactive at
frequencies between 100mhz and 200mhz, and in the 5 to 100 watts range? A
one, basically.

If, by "VSWR meter", you mean the usual directionally sampled line
arrangement, it won't tell you. What it actually measures is forward and
reflected *current*, hence power. Whether the mismatched load is
resistive, capacitive, or inductive, it neither knows nor cares.

Google for "VHF bridge", and see if that gives you any ideas

T

#### tom

Jan 1, 1970
0
I'm not sure I understand your point Mike --- what do you mean by, "Oh,
yeah. Is google's server down or something?"

D

#### dB

Jan 1, 1970
0
tom said:
How minimal of circuit would be required to tell the difference between
whether a load was inductively reactive or capacitively reactive at
frequencies between 100mhz and 200mhz, and in the 5 to 100 watts range? A
one, basically.

A directional power meter (s.w.r. meter) will not differentiate
between the two cases.

N
Replies
6
Views
3K
William Sommerwerck
W
L
Replies
35
Views
3K
B
T
Replies
3
Views
2K
Franc Zabkar
F
D
Replies
5
Views
2K
D
C
Replies
7
Views
2K
Graham
G